Purchase your copy of ASTM D/DM – 10() as a PDF download or hard copy directly from the official BSI Shop. All BSI British. This practice covers the calibration of variable-area flowmeters (rotameters) used to determine air sample volumes at or close to ambient conditions of. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||17 November 2017|
|PDF File Size:||1.3 Mb|
|ePub File Size:||5.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Terminology See Test Method E Use average values and meter water temperature d315 each meter reading.
Please first verify your email before subscribing to alerts. Add to Alert PDF. If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below. Standards Subscriptions from ANSI provides a money-saving, multi-user solution for accessing standards. Since viscosity is propor- added by the wet test meter or gasometer, assuming saturation, tional to the square root of absolute temperature independent by operating with the factor: Already Subscribed to this document.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
ASTM D / DM – 10() Standard Practice for Rotameter Calibration
A number in parentheses indicates the year of last reapproval. Use Methods D Scope ture scales are to be used when substituting values into the 1. Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters.
Please first log in with a verified email before subscribing to alerts.
If the document is revised or amended, you will be notified by email. This standard is not included in any packages. The method described in Section 19 is recommended for Current edition approved Dec. The factor for this correction is: Procedure Using Wet Test Meter 1 7. Proceed to Checkout Continue Shopping. We have no amendments or corrections for this standard. Users of this standard are expressly advised that determination of the validity of any such patent rights, and the risk of infringement of such rights, are entirely their own responsibility.
As the voice of the U. Repeat, going from high to low.
ASTM D3195/D3195M – 10(2015)
This conforms to diaphragm type pump of adequate capacity and a ballast most of the ASTM methods for atmospheric sampling and volume or restrictor to eliminate pulsations. Summary of Practice at or close to ambient conditions of pressure and temperature, in the analysis of atmospheres for pollutant content. You may delete a document from your Alert Profile at any time. Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend.
Therefore, in order to prepare the meter; room temperature for gasometer. Note the manometer reading when room air is used for calibrating gas.
Higher accuracy is possible with the gasom- eous Fuel Atm eter. Combining values from the two systems may result in non-conformance with the standard. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service e-mail ; or through the ASTM website. D315 can download and open this file to your own computer but DRM prevents opening this file on another computer, including a networked server.
Last previous edition D — 73 wstm. Subscription pricing is determined by: See bility of regulatory limitations prior to use. A superscript epsilon indicates an editorial change since the last revision or reapproval.
Need more than one copy?